﻿<!DOCTYPE html PUBLIC "-//W3C//DTD XHTML 1.0 Strict//EN"
    "http://www.w3.org/TR/xhtml1/DTD/xhtml1-strict.dtd">

<html xmlns="http://www.w3.org/1999/xhtml">
<head>
  <meta http-equiv="Content-Type" content="text/html; charset=utf-8" />  
  <title>Prosthetic Mitral Valves: Normal Doppler Echocardiographic Values</title>
  <link href="../style/screen.css" rel="stylesheet" type="text/css" />
  <link href="../style/StyleSheet.css" rel="stylesheet" type="text/css" />
  <link href="style/demo_table.css" rel="stylesheet" type="text/css" />
</head>

<body>
  <div class="container">
    <div class="span-24 last">
      <table id="mvrTable" class="display" summary=
      "Normal Doppler Echocardiographic Values for Prosthetic Mitral Valves">
        <thead>
          <tr>
            <th>Valve</th>

            <th>Type</th>

            <th>Size (mm)</th>

            <th>Peak gradient (mmHg)</th>

            <th>Mean gradient (mmHg)</th>

            <th>Peak Velocity (m/s)</th>

            <th>Pressure 1/2 time (ms)</th>

            <th>EOA (cm2)</th>
          </tr>
        </thead>

        <tbody>
          <tr>
            <td>Biocor</td>

            <td>Stentless bioprosthesis</td>

            <td>27</td>

            <td>13 &plusmn; 1</td>

            <td></td>

            <td></td>

            <td></td>

            <td></td>
          </tr>

          <tr>
            <td>Biocor</td>

            <td>Stentless bioprosthesis</td>

            <td>29</td>

            <td>14 &plusmn; 2.5</td>

            <td></td>

            <td></td>

            <td></td>

            <td></td>
          </tr>

          <tr>
            <td>Biocor</td>

            <td>Stentless bioprosthesis</td>

            <td>31</td>

            <td>11.5 &plusmn; 0.5</td>

            <td></td>

            <td></td>

            <td></td>

            <td></td>
          </tr>

          <tr>
            <td>Biocor</td>

            <td>Stentless bioprosthesis</td>

            <td>33</td>

            <td>12 &plusmn; 0.5</td>

            <td></td>

            <td></td>

            <td></td>

            <td></td>
          </tr>

          <tr>
            <td>Bioflo pericardial</td>

            <td>Stented bioprosthesis</td>

            <td>25</td>

            <td>10 &plusmn; 2</td>

            <td>6.3 &plusmn; 1.5</td>

            <td></td>

            <td></td>

            <td>2 &plusmn; 0.1</td>
          </tr>

          <tr>
            <td>Bioflo pericardial</td>

            <td>Stented bioprosthesis</td>

            <td>27</td>

            <td>9.5 &plusmn; 2.6</td>

            <td>5.4 &plusmn; 1.2</td>

            <td></td>

            <td></td>

            <td>2 &plusmn; 0.3</td>
          </tr>

          <tr>
            <td>Bioflo pericardial</td>

            <td>Stented bioprosthesis</td>

            <td>29</td>

            <td>5 &plusmn; 2.8</td>

            <td>3.6 &plusmn; 1</td>

            <td></td>

            <td></td>

            <td>2.4 &plusmn; 0.2</td>
          </tr>

          <tr>
            <td>Bioflo pericardial</td>

            <td>Stented bioprosthesis</td>

            <td>31</td>

            <td>4.0</td>

            <td>2.0</td>

            <td></td>

            <td></td>

            <td>2.3</td>
          </tr>

          <tr>
            <td>Bjork-Shiley</td>

            <td>Tilting disc</td>

            <td>23</td>

            <td></td>

            <td></td>

            <td>1.7</td>

            <td>115</td>

            <td></td>
          </tr>

          <tr>
            <td>Bjork-Shiley</td>

            <td>Tilting disc</td>

            <td>25</td>

            <td>12 &plusmn; 4</td>

            <td>6 &plusmn; 2</td>

            <td>1.75 &plusmn; 0.38</td>

            <td>99 &plusmn; 27</td>

            <td>1.72 &plusmn; 0.6</td>
          </tr>

          <tr>
            <td>Bjork-Shiley</td>

            <td>Tilting disc</td>

            <td>27</td>

            <td>10 &plusmn; 4</td>

            <td>5 &plusmn; 2</td>

            <td>1.6 &plusmn; 0.49</td>

            <td>89 &plusmn; 28</td>

            <td>1.81 &plusmn; 0.54</td>
          </tr>

          <tr>
            <td>Bjork-Shiley</td>

            <td>Tilting disc</td>

            <td>29</td>

            <td>7.83 &plusmn; 2.93</td>

            <td>2.83 &plusmn; 1.27</td>

            <td>1.37 &plusmn; 0.25</td>

            <td>79 &plusmn; 17</td>

            <td>2.1 &plusmn; 0.43</td>
          </tr>

          <tr>
            <td>Bjork-Shiley</td>

            <td>Tilting disc</td>

            <td>31</td>

            <td>6 &plusmn; 3</td>

            <td>2 &plusmn; 1.9</td>

            <td>1.41 &plusmn; 0.26</td>

            <td>70 &plusmn; 14</td>

            <td>2.2 &plusmn; 0.3</td>
          </tr>

          <tr>
            <td>Bjork-Shiley monostrut</td>

            <td>Tilting disc</td>

            <td>23</td>

            <td></td>

            <td>5.0</td>

            <td>1.9</td>

            <td></td>

            <td></td>
          </tr>

          <tr>
            <td>Bjork-Shiley monostrut</td>

            <td>Tilting disc</td>

            <td>25</td>

            <td>13 &plusmn; 2.5</td>

            <td>5.57 &plusmn; 2.3</td>

            <td>1.8 &plusmn; 0.3</td>

            <td></td>

            <td></td>
          </tr>

          <tr>
            <td>Bjork-Shiley monostrut</td>

            <td>Tilting disc</td>

            <td>27</td>

            <td>12 &plusmn; 2.5</td>

            <td>4.53 &plusmn; 2.2</td>

            <td>1.7 &plusmn; 0.4</td>

            <td></td>

            <td></td>
          </tr>

          <tr>
            <td>Bjork-Shiley monostrut</td>

            <td>Tilting disc</td>

            <td>29</td>

            <td>13 &plusmn; 3</td>

            <td>4.26 &plusmn; 1.6</td>

            <td>1.6 &plusmn; 0.3</td>

            <td></td>

            <td></td>
          </tr>

          <tr>
            <td>Bjork-Shiley monostrut</td>

            <td>Tilting disc</td>

            <td>31</td>

            <td>14 &plusmn; 4.5</td>

            <td>4.9 &plusmn; 1.6</td>

            <td>1.7 &plusmn; 0.3</td>

            <td></td>

            <td></td>
          </tr>

          <tr>
            <td>Carbomedics</td>

            <td>Bileaflet</td>

            <td>23</td>

            <td></td>

            <td></td>

            <td>1.9 &plusmn; 0.1</td>

            <td>126 &plusmn; 7</td>

            <td></td>
          </tr>

          <tr>
            <td>Carbomedics</td>

            <td>Bileaflet</td>

            <td>25</td>

            <td>10.3 &plusmn; 2.3</td>

            <td>3.6 &plusmn; 0.6</td>

            <td>1.3 &plusmn; 0.1</td>

            <td>93 &plusmn; 8</td>

            <td>2.9 &plusmn; 0.8</td>
          </tr>

          <tr>
            <td>Carbomedics</td>

            <td>Bileaflet</td>

            <td>27</td>

            <td>8.79 &plusmn; 3.46</td>

            <td>3.46 &plusmn; 1.03</td>

            <td>1.61 &plusmn; 0.3</td>

            <td>89 &plusmn; 20</td>

            <td>2.9 &plusmn; 0.75</td>
          </tr>

          <tr>
            <td>Carbomedics</td>

            <td>Bileaflet</td>

            <td>29</td>

            <td>8.78 &plusmn; 2.9</td>

            <td>3.39 &plusmn; 0.97</td>

            <td>1.52 &plusmn; 0.3</td>

            <td>88 &plusmn; 17</td>

            <td>2.3 &plusmn; 0.4</td>
          </tr>

          <tr>
            <td>Carbomedics</td>

            <td>Bileaflet</td>

            <td>31</td>

            <td>8.87 &plusmn; 2.34</td>

            <td>3.32 &plusmn; 0.87</td>

            <td>1.61 &plusmn; 0.29</td>

            <td>92 &plusmn; 24</td>

            <td>2.8 &plusmn; 1.14</td>
          </tr>

          <tr>
            <td>Carbomedics</td>

            <td>Bileaflet</td>

            <td>33</td>

            <td>8.8 &plusmn; 2.2</td>

            <td>4.8 &plusmn; 2.5</td>

            <td>1.5 &plusmn; 0.2</td>

            <td>93 &plusmn; 12</td>

            <td></td>
          </tr>

          <tr>
            <td>Carpentier-Edwards</td>

            <td>Stented bioprosthesis</td>

            <td>27</td>

            <td></td>

            <td>6 &plusmn; 2</td>

            <td>1.7 &plusmn; 0.3</td>

            <td>98 &plusmn; 28</td>

            <td></td>
          </tr>

          <tr>
            <td>Carpentier-Edwards</td>

            <td>Stented bioprosthesis</td>

            <td>29</td>

            <td></td>

            <td>4.7 &plusmn; 2</td>

            <td>1.76 &plusmn; 0.27</td>

            <td>92 &plusmn; 14</td>

            <td></td>
          </tr>

          <tr>
            <td>Carpentier-Edwards</td>

            <td>Stented bioprosthesis</td>

            <td>31</td>

            <td></td>

            <td>4.4 &plusmn; 2</td>

            <td>1.54 &plusmn; 0.15</td>

            <td>92 &plusmn; 19</td>

            <td></td>
          </tr>

          <tr>
            <td>Carpentier-Edwards</td>

            <td>Stented bioprosthesis</td>

            <td>33</td>

            <td></td>

            <td>6 &plusmn; 3</td>

            <td></td>

            <td>93 &plusmn; 12</td>

            <td></td>
          </tr>

          <tr>
            <td>Carpentier-Edwards pericardial</td>

            <td>Stented Bioprosthesis</td>

            <td>27</td>

            <td></td>

            <td>3.6</td>

            <td>1.6</td>

            <td>100</td>

            <td></td>
          </tr>

          <tr>
            <td>Carpentier-Edwards pericardial</td>

            <td>Stented Bioprosthesis</td>

            <td>29</td>

            <td></td>

            <td>5.25 &plusmn; 2.36</td>

            <td>1.67 &plusmn; 0.3</td>

            <td>110 &plusmn; 15</td>

            <td></td>
          </tr>

          <tr>
            <td>Carpentier-Edwards pericardial</td>

            <td>Stented Bioprosthesis</td>

            <td>31</td>

            <td></td>

            <td>4.05 &plusmn; 0.83</td>

            <td>1.53 &plusmn; 0.1</td>

            <td>90 &plusmn; 11</td>

            <td></td>
          </tr>

          <tr>
            <td>Carpentier-Edwards pericardial</td>

            <td>Stented Bioprosthesis</td>

            <td>23</td>

            <td></td>

            <td>1.0</td>

            <td>0.8</td>

            <td>80</td>

            <td></td>
          </tr>

          <tr>
            <td>Duromedics</td>

            <td>Bileaflet</td>

            <td>27</td>

            <td>13 &plusmn; 6</td>

            <td>5 &plusmn; 3</td>

            <td>1.61 &plusmn; 0.4</td>

            <td>75 &plusmn; 12</td>

            <td></td>
          </tr>

          <tr>
            <td>Duromedics</td>

            <td>Bileaflet</td>

            <td>29</td>

            <td>10 &plusmn; 4</td>

            <td>3 &plusmn; 1</td>

            <td>1.40 &plusmn; 0.25</td>

            <td>85 &plusmn; 22</td>

            <td></td>
          </tr>

          <tr>
            <td>Duromedics</td>

            <td>Bileaflet</td>

            <td>31</td>

            <td>10.5 &plusmn; 4.33</td>

            <td>3.3 &plusmn; 1.36</td>

            <td>1.38 &plusmn; 0.27</td>

            <td>81 &plusmn; 12</td>

            <td></td>
          </tr>

          <tr>
            <td>Duromedics</td>

            <td>Bileaflet</td>

            <td>33</td>

            <td>11.2</td>

            <td>2.5</td>

            <td></td>

            <td>85</td>

            <td></td>
          </tr>

          <tr>
            <td>Hancock I (or <acronym title="Not Otherwise Specified">NOS</acronym>)</td>

            <td>Stented bioprosthesis</td>

            <td>27</td>

            <td>10 &plusmn; 4</td>

            <td>5 &plusmn; 2</td>

            <td></td>

            <td></td>

            <td>1.3 &plusmn; 0.8</td>
          </tr>

          <tr>
            <td>Hancock I (or <acronym title="Not Otherwise Specified">NOS</acronym>)</td>

            <td>Stented bioprosthesis</td>

            <td>29</td>

            <td>7 &plusmn; 3</td>

            <td>2.46 &plusmn; 0.79</td>

            <td></td>

            <td>115 &plusmn; 20</td>

            <td>1.5 &plusmn; 0.2</td>
          </tr>

          <tr>
            <td>Hancock I (or <acronym title="Not Otherwise Specified">NOS</acronym>)</td>

            <td>Stented bioprosthesis</td>

            <td>31</td>

            <td>4 &plusmn; 0.86</td>

            <td>4.86 &plusmn; 1.69</td>

            <td></td>

            <td>95 &plusmn; 17</td>

            <td>1.6 &plusmn; 0.2</td>
          </tr>

          <tr>
            <td>Hancock I (or <acronym title="Not Otherwise Specified">NOS</acronym>)</td>

            <td>Stented bioprosthesis</td>

            <td>33</td>

            <td>3 &plusmn; 2</td>

            <td>3.87 &plusmn; 2</td>

            <td></td>

            <td>90 &plusmn; 12</td>

            <td>1.9 &plusmn; 0.2</td>
          </tr>

          <tr>
            <td>Hancock II</td>

            <td>Stented bioprosthesis</td>

            <td>27</td>

            <td></td>

            <td></td>

            <td></td>

            <td></td>

            <td>2.21 &plusmn; 0.14</td>
          </tr>

          <tr>
            <td>Hancock II</td>

            <td>Stented bioprosthesis</td>

            <td>29</td>

            <td></td>

            <td></td>

            <td></td>

            <td></td>

            <td>2.77 &plusmn; 0.11</td>
          </tr>

          <tr>
            <td>Hancock II</td>

            <td>Stented bioprosthesis</td>

            <td>31</td>

            <td></td>

            <td></td>

            <td></td>

            <td></td>

            <td>2.84 &plusmn; 0.1</td>
          </tr>

          <tr>
            <td>Hancock II</td>

            <td>Stented bioprosthesis</td>

            <td>33</td>

            <td></td>

            <td></td>

            <td></td>

            <td></td>

            <td>3.15 &plusmn; 0.22</td>
          </tr>

          <tr>
            <td>Hancock pericardial</td>

            <td>Stented bioprosthesis</td>

            <td>29</td>

            <td></td>

            <td>2.61 &plusmn; 1.39</td>

            <td>1.42 &plusmn; 0.14</td>

            <td>105 &plusmn; 36</td>

            <td></td>
          </tr>

          <tr>
            <td>Hancock pericardial</td>

            <td>Stented bioprosthesis</td>

            <td>31</td>

            <td></td>

            <td>3.57 &plusmn; 1.02</td>

            <td>1.51 &plusmn; 0.27</td>

            <td>81 &plusmn; 23</td>

            <td></td>
          </tr>

          <tr>
            <td>Ionescu-Shiley</td>

            <td>Stented bioprosthesis</td>

            <td>25</td>

            <td></td>

            <td>4.87 &plusmn; 1.08</td>

            <td>1.43 &plusmn; 0.15</td>

            <td>93 &plusmn; 11</td>

            <td></td>
          </tr>

          <tr>
            <td>Ionescu-Shiley</td>

            <td>Stented bioprosthesis</td>

            <td>27</td>

            <td></td>

            <td>3.21 &plusmn; 0.82</td>

            <td>1.31 &plusmn; 0.24</td>

            <td>100 &plusmn; 28</td>

            <td></td>
          </tr>

          <tr>
            <td>Ionescu-Shiley</td>

            <td>Stented bioprosthesis</td>

            <td>29</td>

            <td></td>

            <td>3.22 &plusmn; 0.57</td>

            <td>1.38 &plusmn; 0.2</td>

            <td>85 &plusmn; 8</td>

            <td></td>
          </tr>

          <tr>
            <td>Ionescu-Shiley</td>

            <td>Stented bioprosthesis</td>

            <td>31</td>

            <td></td>

            <td>3.63 &plusmn; 0.9</td>

            <td>1.45 &plusmn; 0.06</td>

            <td>100 &plusmn; 36</td>

            <td></td>
          </tr>

          <tr>
            <td>Ionescu-Shiley low profile</td>

            <td>Stented bioprosthesis</td>

            <td>29</td>

            <td></td>

            <td>3.31 &plusmn; 0.96</td>

            <td>1.36 &plusmn; 0.25</td>

            <td>80 &plusmn; 30</td>

            <td></td>
          </tr>

          <tr>
            <td>Ionescu-Shiley low profile</td>

            <td>Stented bioprosthesis</td>

            <td>31</td>

            <td></td>

            <td>2.74 &plusmn; 0.37</td>

            <td>1.33 &plusmn; 0.14</td>

            <td>79 &plusmn; 15</td>

            <td></td>
          </tr>

          <tr>
            <td>Labcor-Santiago pericardial</td>

            <td>Stented bioprosthesis</td>

            <td>25</td>

            <td>8.7</td>

            <td>4.5</td>

            <td></td>

            <td>97</td>

            <td>2.2</td>
          </tr>

          <tr>
            <td>Labcor-Santiago pericardial</td>

            <td>Stented bioprosthesis</td>

            <td>27</td>

            <td>5.6 &plusmn; 2.3</td>

            <td>2.8 &plusmn; 1.5</td>

            <td></td>

            <td>85 &plusmn; 18</td>

            <td>2.12 &plusmn; 0.48</td>
          </tr>

          <tr>
            <td>Labcor-Santiago pericardial</td>

            <td>Stented bioprosthesis</td>

            <td>29</td>

            <td>6.2 &plusmn; 2.1</td>

            <td>3 &plusmn; 1.3</td>

            <td></td>

            <td>80 &plusmn; 34</td>

            <td>2.11 &plusmn; 0.73</td>
          </tr>

          <tr>
            <td>Lillehei-Kaster</td>

            <td>Tilting disc</td>

            <td>18</td>

            <td></td>

            <td></td>

            <td>1.7</td>

            <td>140</td>

            <td></td>
          </tr>

          <tr>
            <td>Lillehei-Kaster</td>

            <td>Tilting disc</td>

            <td>20</td>

            <td></td>

            <td></td>

            <td>1.7</td>

            <td>67</td>

            <td></td>
          </tr>

          <tr>
            <td>Lillehei-Kaster</td>

            <td>Tilting disc</td>

            <td>22</td>

            <td></td>

            <td></td>

            <td>1.56 &plusmn; 0.09</td>

            <td>94 &plusmn; 22</td>

            <td></td>
          </tr>

          <tr>
            <td>Lillehei-Kaster</td>

            <td>Tilting disc</td>

            <td>25</td>

            <td></td>

            <td></td>

            <td>1.38 &plusmn; 0.27</td>

            <td>124 &plusmn; 46</td>

            <td></td>
          </tr>

          <tr>
            <td>Medtronic Hall</td>

            <td>Tilting disc</td>

            <td>27</td>

            <td></td>

            <td></td>

            <td>1.4</td>

            <td>78</td>

            <td></td>
          </tr>

          <tr>
            <td>Medtronic Hall</td>

            <td>Tilting disc</td>

            <td>29</td>

            <td></td>

            <td></td>

            <td>1.57 &plusmn; 0.1</td>

            <td>69 &plusmn; 15</td>

            <td></td>
          </tr>

          <tr>
            <td>Medtronic Hall</td>

            <td>Tilting disc</td>

            <td>31</td>

            <td></td>

            <td></td>

            <td>1.45 &plusmn; 0.12</td>

            <td>77 &plusmn; 17</td>

            <td></td>
          </tr>

          <tr>
            <td>Medtronic Intact Porcine</td>

            <td>Stented bioprosthesis</td>

            <td>29</td>

            <td></td>

            <td>3.5 &plusmn; 0.51</td>

            <td>1.6 &plusmn; 0.22</td>

            <td></td>

            <td></td>
          </tr>

          <tr>
            <td>Medtronic Intact Porcine</td>

            <td>Stented bioprosthesis</td>

            <td>31</td>

            <td></td>

            <td>4.2 &plusmn; 1.44</td>

            <td>1.6 &plusmn; 0.26</td>

            <td></td>

            <td></td>
          </tr>

          <tr>
            <td>Medtronic Intact Porcine</td>

            <td>Stented bioprosthesis</td>

            <td>33</td>

            <td></td>

            <td>4 &plusmn; 1.3</td>

            <td>1.4 &plusmn; 0.24</td>

            <td></td>

            <td></td>
          </tr>

          <tr>
            <td>Medtronic Intact Porcine</td>

            <td>Stented bioprosthesis</td>

            <td>35</td>

            <td></td>

            <td>3.2 &plusmn; 1.77</td>

            <td>1.3 &plusmn; 0.5</td>

            <td></td>

            <td></td>
          </tr>

          <tr>
            <td>Mitroflow</td>

            <td>Stented bioprosthesis</td>

            <td>25</td>

            <td></td>

            <td>6.9</td>

            <td>2.0</td>

            <td>90</td>

            <td></td>
          </tr>

          <tr>
            <td>Mitroflow</td>

            <td>Stented bioprosthesis</td>

            <td>27</td>

            <td></td>

            <td>3.07 &plusmn; 0.91</td>

            <td>1.5</td>

            <td>90 &plusmn; 20</td>

            <td></td>
          </tr>

          <tr>
            <td>Mitroflow</td>

            <td>Stented bioprosthesis</td>

            <td>29</td>

            <td></td>

            <td>3.5 &plusmn; 1.65</td>

            <td>1.43 &plusmn; 0.29</td>

            <td>102 &plusmn; 21</td>

            <td></td>
          </tr>

          <tr>
            <td>Mitroflow</td>

            <td>Stented bioprosthesis</td>

            <td>31</td>

            <td></td>

            <td>3.85 &plusmn; 0.81</td>

            <td>1.32 &plusmn; 0.26</td>

            <td>91 &plusmn; 22</td>

            <td></td>
          </tr>

          <tr>
            <td>Omnicarbon</td>

            <td>Tilting disc</td>

            <td>23</td>

            <td></td>

            <td>8.0</td>

            <td></td>

            <td></td>

            <td></td>
          </tr>

          <tr>
            <td>Omnicarbon</td>

            <td>Tilting disc</td>

            <td>25</td>

            <td></td>

            <td>6.05 &plusmn; 1.81</td>

            <td>1.77 &plusmn; 0.24</td>

            <td>102 &plusmn; 16</td>

            <td></td>
          </tr>

          <tr>
            <td>Omnicarbon</td>

            <td>Tilting disc</td>

            <td>27</td>

            <td></td>

            <td>4.89 &plusmn; 2.05</td>

            <td>1.63 &plusmn; 0.36</td>

            <td>105 &plusmn; 33</td>

            <td></td>
          </tr>

          <tr>
            <td>Omnicarbon</td>

            <td>Tilting disc</td>

            <td>29</td>

            <td></td>

            <td>4.93 &plusmn; 2.16</td>

            <td>1.56 &plusmn; 0.27</td>

            <td>120 &plusmn; 40</td>

            <td></td>
          </tr>

          <tr>
            <td>Omnicarbon</td>

            <td>Tilting disc</td>

            <td>31</td>

            <td></td>

            <td>4.18 &plusmn; 1.4</td>

            <td>1.3 &plusmn; 0.23</td>

            <td>134 &plusmn; 31</td>

            <td></td>
          </tr>

          <tr>
            <td>Omnicarbon</td>

            <td>Tilting disc</td>

            <td>33</td>

            <td></td>

            <td>4 &plusmn; 2</td>

            <td></td>

            <td></td>

            <td></td>
          </tr>

          <tr>
            <td>On-X</td>

            <td>Bileaflet</td>

            <td>25</td>

            <td>11.5 &plusmn; 3.2</td>

            <td>5.3 &plusmn; 2.1</td>

            <td></td>

            <td></td>

            <td>1.9 &plusmn; 1.1</td>
          </tr>

          <tr>
            <td>On-X</td>

            <td>Bileaflet</td>

            <td>27</td>

            <td>10.3 &plusmn; 4.5</td>

            <td>4.5 &plusmn; 1.6</td>

            <td></td>

            <td></td>

            <td>2.2 &plusmn; 0.5</td>
          </tr>

          <tr>
            <td>On-X</td>

            <td>Bileaflet</td>

            <td>29</td>

            <td>10.3 &plusmn; 4.5</td>

            <td>4.5 &plusmn; 1.6</td>

            <td></td>

            <td></td>

            <td>2.2 &plusmn; 0.5</td>
          </tr>

          <tr>
            <td>On-X</td>

            <td>Bileaflet</td>

            <td>31</td>

            <td>9.8 &plusmn; 3.8</td>

            <td>4.8 &plusmn; 2.4</td>

            <td></td>

            <td></td>

            <td>2.5 &plusmn; 1.1</td>
          </tr>

          <tr>
            <td>On-X</td>

            <td>Bileaflet</td>

            <td>33</td>

            <td>9.8 &plusmn; 3.8</td>

            <td>4.8 &plusmn; 2.4</td>

            <td></td>

            <td></td>

            <td>2.5 &plusmn; 1.1</td>
          </tr>

          <tr>
            <td>Sorin Allcarbon</td>

            <td>Tilting disc</td>

            <td>25</td>

            <td>15 &plusmn; 0.25</td>

            <td>4 &plusmn; 0.5</td>

            <td>1.95 &plusmn; 0.02</td>

            <td>70 &plusmn; 1</td>

            <td></td>
          </tr>

          <tr>
            <td>Sorin Allcarbon</td>

            <td>Tilting disc</td>

            <td>27</td>

            <td>11 &plusmn; 2.75</td>

            <td>4 &plusmn; 0.5</td>

            <td>1.65 &plusmn; 0.21</td>

            <td>82 &plusmn; 20</td>

            <td></td>
          </tr>

          <tr>
            <td>Sorin Allcarbon</td>

            <td>Tilting disc</td>

            <td>29</td>

            <td>12 &plusmn; 3</td>

            <td>4 &plusmn; 1.25</td>

            <td>1.73 &plusmn; 0.22</td>

            <td>80 &plusmn; 14</td>

            <td></td>
          </tr>

          <tr>
            <td>Sorin Allcarbon</td>

            <td>Tilting disc</td>

            <td>31</td>

            <td>10 &plusmn; 1.5</td>

            <td>4 &plusmn; 1</td>

            <td>1.66 &plusmn; 0.11</td>

            <td>83 &plusmn; 14</td>

            <td></td>
          </tr>

          <tr>
            <td>St. Jude Medical</td>

            <td>Bileaflet</td>

            <td>23</td>

            <td></td>

            <td>4.0</td>

            <td>1.5</td>

            <td>160</td>

            <td>1.0</td>
          </tr>

          <tr>
            <td>St. Jude Medical</td>

            <td>Bileaflet</td>

            <td>25</td>

            <td></td>

            <td>2.5 &plusmn; 1</td>

            <td>1.34 &plusmn; 1.12</td>

            <td>75 &plusmn; 4</td>

            <td>1.35 &plusmn; 0.17</td>
          </tr>

          <tr>
            <td>St. Jude Medical</td>

            <td>Bileaflet</td>

            <td>27</td>

            <td>11 &plusmn; 4</td>

            <td>5 &plusmn; 1.82</td>

            <td>1.61 &plusmn; 0.29</td>

            <td>75 &plusmn; 10</td>

            <td>1.67 &plusmn; 0.17</td>
          </tr>

          <tr>
            <td>St. Jude Medical</td>

            <td>Bileaflet</td>

            <td>29</td>

            <td>10 &plusmn; 3</td>

            <td>4.15 &plusmn; 1.8</td>

            <td>1.57 &plusmn; 0.29</td>

            <td>85 &plusmn; 10</td>

            <td>1.75 &plusmn; 0.24</td>
          </tr>

          <tr>
            <td>St. Jude Medical</td>

            <td>Bileaflet</td>

            <td>31</td>

            <td>12 &plusmn; 6</td>

            <td>4.46 &plusmn; 2.22</td>

            <td>1.59 &plusmn; 0.33</td>

            <td>74 &plusmn; 13</td>

            <td>2.03 &plusmn; 0.32</td>
          </tr>

          <tr>
            <td>Starr-Edwards</td>

            <td>Caged ball</td>

            <td>26</td>

            <td></td>

            <td>10.0</td>

            <td></td>

            <td></td>

            <td>1.4</td>
          </tr>

          <tr>
            <td>Starr-Edwards</td>

            <td>Caged ball</td>

            <td>28</td>

            <td></td>

            <td>7 &plusmn; 2.75</td>

            <td></td>

            <td></td>

            <td>1.9 &plusmn; 0.57</td>
          </tr>

          <tr>
            <td>Starr-Edwards</td>

            <td>Caged ball</td>

            <td>30</td>

            <td>12.2 &plusmn; 4.6</td>

            <td>6.99 &plusmn; 2.5</td>

            <td>1.7 &plusmn; 0.3</td>

            <td>125 &plusmn; 25</td>

            <td>1.65 &plusmn; 0.4</td>
          </tr>

          <tr>
            <td>Starr-Edwards</td>

            <td>Caged ball</td>

            <td>32</td>

            <td>11.5 &plusmn; 4.2</td>

            <td>5.08 &plusmn; 2.5</td>

            <td>1.7 &plusmn; 0.3</td>

            <td>110 &plusmn; 25</td>

            <td>1.98 &plusmn; 0.4</td>
          </tr>

          <tr>
            <td>Starr-Edwards</td>

            <td>Caged ball</td>

            <td>34</td>

            <td></td>

            <td>5.0</td>

            <td></td>

            <td></td>

            <td>2.6</td>
          </tr>

          <tr>
            <td>Stentless quadrileaflet bovine pericardial</td>

            <td>Stentless bioprosthesis</td>

            <td>26</td>

            <td></td>

            <td>2.2 &plusmn; 1.7</td>

            <td>1.6</td>

            <td>103 &plusmn; 31</td>

            <td>1.7</td>
          </tr>

          <tr>
            <td>Stentless quadrileaflet bovine pericardial</td>

            <td>Stentless bioprosthesis</td>

            <td>28</td>

            <td></td>

            <td></td>

            <td>1.58 &plusmn; 0.25</td>

            <td></td>

            <td>1.7 &plusmn; 0.6</td>
          </tr>

          <tr>
            <td>Stentless quadrileaflet bovine pericardial</td>

            <td>Stentless bioprosthesis</td>

            <td>30</td>

            <td></td>

            <td></td>

            <td>1.42 &plusmn; 0.32</td>

            <td></td>

            <td>2.3 &plusmn; 0.4</td>
          </tr>

          <tr>
            <td>Wessex</td>

            <td>Stented bioprosthesis</td>

            <td>29</td>

            <td></td>

            <td>3.69 &plusmn; 0.61</td>

            <td>1.66 &plusmn; 0.17</td>

            <td>83 &plusmn; 19</td>

            <td></td>
          </tr>

          <tr>
            <td>Wessex</td>

            <td>Stented bioprosthesis</td>

            <td>31</td>

            <td></td>

            <td>3.31 &plusmn; 0.83</td>

            <td>1.41 &plusmn; 0.25</td>

            <td>80 &plusmn; 21</td>

            <td></td>
          </tr>
        </tbody>
      </table>
    </div><!-- end main div -->
  </div><!-- end container div -->
  <script src="http://ajax.googleapis.com/ajax/libs/jquery/1/jquery.min.js" type=
  "text/javascript">
</script><script src="js/jquery.dataTables.min.js" type="text/javascript">
</script><script type="text/javascript">
             $(document).ready(function() {
                 $('#mvrTable').dataTable({
                     "fnRowCallback": function(nRow, aData, iDisplayIndex) {
                         /* Append the grade to the default row class name */
                         if (aData[1] == "Bileaflet") {
                             $(nRow).addClass('Bileaflet');
                         }
                         return nRow;
                     },
                     "aoColumns": [
        { "sWidth": "18em" }, //valve
        { "sWidth": "12em" }, //type
        null, //size
        { "bSearchable": false }, //peakGr
        { "bSearchable": false }, //meanGr
        { "bSearchable": false }, //peakV
        { "bSearchable": false }, //pr1/2t
        { "bSearchable": false} //eoa
        ],
                     "bPaginate": false

                 });
                 $('#mvrTable_filter input').val('Example: 23 bov')
            .addClass('search_init')
            .focus(function() {
                if (this.className == "search_init") {
                    this.className = "";
                    this.value = "";
                }
            })
            .blur(function() {
                if (this.value == "") {
                    this.className = "search_init";
                    this.value = "Example: 23 bov";
                }
            });
             });
  </script>
</body>
</html>
